Search the Community
Showing results for tags 'overcurrent'.
Found 2 results
Hi, I want to use NI-9227 on NI-cDAQ 9188 chassis for current measurement. When the load is engaged an instantaneous current of 25 Amps for microseconds is observed using Oscilloscope. The specifications of this card says, Input Impedance = 12mOhms (across the channels Ai+ and Ai-) and Instantaneous Measuring Range is 14 Amps. My concern is whether this card will still support this in-rush current or the channel will go bad or the card will not work ? Any suggestion is welcomed. Regards, Runjhun.