Search the Community
Showing results for tags 'ni-9237'.
Hello everyone i am relative new to labview and currently i am bit stuck need help... i want to capture data with a digital trigger on my chassis PFI0 (9178), the pulse is 20ms long and occurs on my straingauge 9237 module. The pulse comes in 10 times in a row with couple of miliseconds between it, can i measure it every time and push it inside an array like a "last in first out" memmory to evaluate it slower for the user? I am currently struggeling with the loop dividing into more then one loop approach. How to breach out from accquire->store->pl
Have configured my NI-9237 as an AI bridge, reading in CONTINUOUS with a very low sample rate and SAMPLES at ONE. I'm running on a FAST Dell Precision T7500 with USB to a cDAQ-9178. But I still get warnings of memory overrun due to not being able to read data fast enough. Tried a bunch of other values and none of them work. I rebooted the PC and the cDAQ and still no go. Any ideas?