Search the Community
Showing results for tags 'ni max'.
Hello fellow LabVIEWers, I am trying to get a self developed PXI chassis up and running - with some problems. As this is this my first time developing a chassis, some general questions about 3rd party chassis: If I want to get a 3rd party PXI chassis up and running with MAX I just need to add the right description files into the right folder? otherwise the chassis will be recognized as "undefined"? but will still work - except triggers? Whats my status right now? Chassis is connected via PCIe 8361 - PXIe 8360 and the Connection seems to work
I would like to know if there is a parameter in line scan camera which allow me to control time trigger along NI MAX or labview with imaqdx. I mean, I would like to triggering camera during 100ms which specific shutter and specific line/sec. And if I change shutter or line/sec the time trigger keep being the same (100ms) I don't know if this is possible with some parameter of line scan camera. Thanks a lot.
Hello to all. I am using NI MAX to get images from a camera (snao mode and hardware trigger). The camera is a basler 1300-30gm. The problem is the next: My camera snaps 4 images (1 image each 40ms). In pylon viewer (camera oficcial software) I can see the four images correctly. The problem is that when I am trying to see the 4 images in NI MAX sometime the camera can not take the 4th image... like there is no time to take the 4 image (or process it to visualize in NI MAX). So maybe there is a configuration parameter in NI MAX to solve this. Any ideas? Ho